- Original Article
- Open Access
The skin dose of pelvic radiographs since 1896
© The Author(s). 2019
- Received: 21 November 2018
- Accepted: 23 January 2019
- Published: 29 March 2019
To derive conversions of antiquated exposure data into modern equivalents and to apply these in the assessment of the skin dose of pelvic radiographs since 1896.
The literature 1896–2018 was searched for implicit and explicit dose information. The early implicit dose data contained now obsolete descriptions of radiation quality and quantity for long since disappeared X-ray systems of limited efficiency. Converting the old information into modern specifications was achieved using contemporary data and computer simulations. Final dose calculations were done with modern software. Explicit radiation doses of later date reported in old quantities and units were adapted according to current recommendations.
For the period before 1927 conversion algorithms for spark gap data and penetrometer hardnesses to high voltage could be derived. Electrical and X-ray efficiencies of several old röntgen systems were determined. Together they allowed reconstruction of 53 doses. After 1927 doses were generally explicitly specified; 114 were retrieved. Although an enormous spread was observed, the average skin dose was reduced by a factor of about 400.
Antiquated exposure data were successfully used for dose reconstruction. Extreme dose variability was a constant. Efforts to cut down doses were effective as skin doses went down from sub-erythema values to about one milligray.
- History radiology
- First X-ray systems
- Early dosimetry
- Pelvic radiography
- Dose reconstruction
Implicit, antiquated dose information from 1896 to 1926 was successfully used in dose reconstruction
Explicit doses appeared after 1927, but were up to recent times in now outdated quantities and units
During 1896–2018 the dose of pelvic radiographs was reduced by a factor of about 400
At all times an extremely large spread in reconstructed and retrieved doses was observed
Dose reconstruction methods developed in this study might be applied to radiography or fluoroscopy of other body parts
Investigating the origin of gonad shielding, we searched for information on radiation doses around 1900. It turned out that such data are effectively missing, probably because dosimetry, as we know it today, was only introduced around 1927. To our knowledge only Kotre et al. tried to estimate radiation doses before this time in a quantitative way, actually for patients and staff in Newcastle between 1899 and 1902 . We decided to address this issue on a larger scale, taking the entrance dose required for a radiograph in the pelvic area as the main subject of our investigation. In this study information from the literature was combined with results from computer simulations.
With hindsight the dosimetric problems of users of X-ray systems directly after 1895 are easy to see. Their theoretical framework was incomplete, they lacked the tools to measure all essential system characteristics and their X-ray tubes were inconstant in their functioning. It explains why users gave no (or at best incomplete) descriptions of X-ray exposures. Dose reconstruction for this period is therefore only possible if missing information can be deduced from other sources. Between 1900 and 1910 descriptions of radiographic exposures improved from deficient to complete. However, until 1927 information on high voltage, radiation hardness and dose was always in antiquated form. The transformation of these old data to modern quantities, including the assessment of the röntgen systems involved, is at the core of this study. To put the newly estimated doses from early radiology into perspective, explicit dose data from 1927 through 2018 were retrieved and converted into current equivalents if necessary.
To limit the size of the text, most background information and results from simulations were moved to Additional file 1, often already referred to in the header of a section. In principle the main text can be read without consulting these more technical addenda.
Dose information on pelvic radiography for the period 1896–2018 was collected from scientific journals, books and documentation from manufacturers of X-ray equipment in English, German and French. To reduce selection bias, any consulted publication containing adequate dose information was included. If for some parameter a range was given, we took the middle of that range. When in later years the dose quantity (KfiA, ESAK, SAD) was not specified, we would assume what appeared most probable. Likewise, when in later implicit dose data the filtration was missing, we chose a value typical for the time.
Reconstruction of doses before 1927 – the difficult but interesting period
Chromoradiometers (Additional file 1: ES I)
Between 1902 and 1907 several dosimeters were introduced that relied on a chemical or physical change of some substance under the influence of X-rays (e.g. color change of a salt or sensitization of a photographic emulsion). These so-called chromoradiometers were calibrated with a dose corresponding to a certain radiation induced affection of the skin. Relevant to us is the epilation or “pastille” dose amounting to about 3.9 Gy . With this calibration old readings for clinical exposures can easily be converted into ESAK. Unfortunately radiometer data were scarce for diagnostic radiology, so our dose reconstruction had mainly to rely on electrical, geometrical and possibly other information that characterizes an X-ray exposure. Dose estimation along this line we call the “computational approach”.
Identifying the X-rays systems of early radiography associated with useful exposure data.
Determining the parameters relevant for the calculation of the dose of a radiograph, and if these were specified in antiquated form, establishing the conversion into modern quantities and units.
Finding the output of the ancient (pulsed) X-ray systems relative to that of a direct current (DC) system. This is necessary because modern software programs required for the calculation of dose and the simulation of X-ray spectra are available for DC systems only. From such programs we selected the powerful and user friendly SpekCalc .
Retrieving clinical exposure data and computing skin doses.
Steps I-III are presented here under “Materials and Methods”, step IV in the “Results” section.
In our computations we relied on the NIST attenuation data [6, 7] and the “anti-plotter” program Graph Grabber . Additional information on software and other resources is given in Additional file 1: ES II.
Step I computational approach. Relevant ancient X-ray systems (Additional file 1: ES III and ES IV)
A. Inductor coil with interrupter and ion tube (1896+)
B. Transformer, mostly with mechanical rectifier (‘Snook’, 1907+), and ion tube
C1. Transformer with mechanical rectifier (Snook) and Coolidge tube (1913+)
C2. Transformer and self-rectifying Coolidge tube (1913+), or transformer with vacuum tube rectifiers and Coolidge tube (1926+)
D. Inductor and Coolidge tube (1913+)
All systems mentioned applied pulsed voltage to the X-ray tube. We will see that C1 and C2 gave similar output and radiation quality. Additional characteristics of these systems can be found in Additional file 1: ES III; for thermal properties of some early ion tubes see Additional file 1: ES IV.
Step II computational approach. The parameters that are required
Parameters relevant in characterizing historical radiographs
Old information (pre-1927)
Peak high-voltage (kVp)
Equivalent spark gap on functioning tube
Hardness X-rays according to penetrometer or half value layer in water
Electrical information on primary and secondary circuit
Tube current (mA)a
Average current, after 1904 often specified
Electrical information on primary and secondary circuit
Exposure time (min, s)a
Explicitly specified duration of exposure
Target (anode) material
Explicit or implicit: before 1912 platinum (Pt), then replacement by tungsten (W)
Implicit: no filter added, i.e. tube envelope only. Later sometimes specified filter
Focus-plate or focus-patient distance, field size, type of study
kVp-info, filtration, object thickness, expo-geometry. Explicit BSF seldom & later
We now consider the parameters individually.
– kVp – from spark gap (Additional file 1 : ES V)
The high voltage (HV) over the X-ray tube is the main determinant of the radiation quality (penetrating power) and an important factor in the X-ray output rate (intensity). In early radiology it proved impossible to measure a good proxy for the pulsed voltages commonly used. X-ray users resorted therefore to measuring quantities that were related to it. Best known is the spark gap, where this name can both refer to the length a spark could bridge and the hardware for its measurement (two pins, a pin and plate, or two spheres).
Measuring the spark gap was simple, but it appeared that the quality of the X-ray output could vary between systems even if the spark gaps were the same. Possible causes of these disparities are discussed in Additional file 1: ES V.
– kVp – from penetrometer hardness (Additional file 1 : ES VI)
Several other penetrometers came into use as well. We consider only the devices encountered in descriptions of exposures, which were in addition to Benoist’s those according Benoist-Walter, Wehnelt and Walter. For the latter three relations between hardness and spark gap are shown in Additional file 1: ES VI. SpekCalc simulations of the Benoist and Wehnelt penetrometers are also presented there.
Christen provided a major contribution to dosimetry in 1913 with his monograph “Messung und Dosierung der Röntgenstrahlen” . He advocated the use of the half value layer (HVL) of X-rays in water for quantifying their hardness. However, we found virtually no HVL data for clinical radiography, but data on the relation between hardness and HVL were relatively numerous; these are also shown in Additional file 1: ES VI.
– kVp – from electrical information (Additional file 1 : ES VII)
Electrical transformation efficiency (α) of inductor with various interrupters (Additional file 1: ES VII)
Transformation efficiency α
Incidentally, for the soft radiation of early radiology simulations show, for a fixed secondary power, that the kVp and tube current can be varied within large margins without changing the KfiA (see Additional file 1: Figure ES13). This means that any (reasonable) kVp can be chosen to calculate the tube current from Psec. However, if given, the reported kVp was used.
– mAs or mA.min
Contrary to the high voltage over the tube, the current through the tube could relatively easily be measured with a moving coil milliampere meter, introduced in 1904 by Gaiffe . Exposure time was initially measured and controlled manually and later (electro-) mechanically.
– Anode material – Pt, W (Additional file 1 : ES VIII)
The first dose data pertained to systems with ion tubes having a platinum (Pt) anode (called the anti-cathode in tubes with three electrodes). After 1912 tungsten (W) started to replace platinum. Coolidge also used W for the anode in his hot cathode tube from 1913 . Because the X-ray output is approximately proportional to the atomic number Z, output correction for instance from W (as assumed in SpekCalc) to the actually used Pt is easy: multiplication of the W-output by ZPt/ZW = 78/74 = 1.054 (the anode factor). The anode angle in ion tubes, both for gas and Coolidge tubes, was typically 45 degrees until the introduction of the line (“Goetze”) focus in 1922 .
– Filtration – glass envelope of tube, additional filter, air (Additional file 1 : ES IX)
Several early authors gave information on the thickness of the tube’s glass wall in the exit region of the X-rays; they all agree on values between 0.2 and 1.1 mm, the actual value being related to the diameter of the tube. In our calculations we assumed a wall thickness of 0.85 mm glass, except for two small ion tubes from 1896, for which we took 0.6 mm (Additional file 1: ES IX). For a very long time walls remained thin and until the twenties generally no additional filter was applied in radiography.
– Exposure geometry
In descriptions of the geometry used in making a pelvic radiograph often only the distance between the X-ray focus and photographic plate was specified. In such cases we assumed a patient thickness of 20 cm and a distance of 1 cm between patient and plate. When in later years (1920+) the use of a Bucky-Potter grid was reported the latter distance was set to 5 cm.
– Backscattered X-rays
The backscatter factor (BSF) mainly depends on high voltage, X-ray filtration, field size and object thickness, and was calculated from data published by Harrison  and the NCRP Report 68 ; most published BSF-values apply to too heavily filtered X-rays. For pelvic radiography, BSF-values varied from 1.11 to 1.17 (after 1927 up to 1.49).
Step III computational approach. Estimation of output of a pulsed system from that of a DC system (Additional file 1: ES X)
After Step II we have, for a given röntgen system, information on all exposure parameters specified in Table 1. For dose calculation using a “DC program” we now have to relate the output of a tube excited with a pulsed voltage to one driven by a DC power supply. One way of accomplishing this is by determining an “X-ray efficiency factor” (ε) with which the output of a DC system, using the kVp of the pulsed system, must be scaled to get the output of the pulsed system. The factor ε we are looking for depends on the waveforms of both the voltage and current exciting the X-ray tube, which vary with the type of the HV-generator and X-ray tube.
Different combinations of HV-generator and X-ray tube and their X-ray efficiency ε
Ratio (KfiA pulsed)/(KfiA DC) (ε)
The KfiA of some old system, applying a pulsed high-voltage of kVp, can now be obtained by performing a SpekCalc (i.e. DC) simulation at the given value kVp and multiplying the KfiA from the simulation with the corresponding factor ε from Table 3. Multiplication with the BSF, and possibly the anode factor (ZPt/ZW), yields the ESAK. An overview of all data used in determining ε is given in Additional file 1: ES XI, whereas Additional file 1: ESXII estimates the uncertainty in the reconstructed dose.
Our search for input for dose reconstruction revealed the impossibility of early radiology to make high quality anteroposterior radiographs of the (large) adult pelvis, mainly due to the loss of contrast by excessive scattered radiation. Good small-area X-ray radiographs of the hip, lumbar spine, kidney stones and abdomen were feasible though. These required more or less similar radiation quality and dose as the full pelvis. For the early years we used therefore information on all four types of images. Good radiographs of the adult pelvis could be made following the introduction of the moving Bucky-Potter anti-scatter grid after 1920 .
Retrieval of “modern” dose data after 1927 – the easy period
Conversions of old quantities into ‘kerma in air’ or ‘absorbed dose’a
Kerma in air
Results from computational approach 1896–1900
Directly after 1895 information on X-ray exposures was always incomplete. However, if we interpret the various descriptions of the glowing anode during radiography as a quantitative indication of the temperature, X-ray dose estimation comes within reach. Assuming an anode temperature of 750 °C (Additional file 1: ES IV), the maximum electrical loads of two popular tubes, the Jackson focus tube and one according to Chabaud, were estimated to be roughly 5 and 10 W, respectively.
Some exposure parameters from before 1900 and the corresponding ESAK
Sensitivity of the KfiA to basic parameters for an 1896 X-ray tube with glowing anode
Rosenfeld (1897) 
Total change KfiA towards
Kotre et al.
Kotre et al. performed for the period 1899–1902 a nice replication study using old equipment and logbooks with exposure data from the Forth Banks Infirmary in Newcastle. They reported a median ESAK of 189 mGy; according to their (skewed) Fig. 4 the average was about 213 mGy .
Chromoradiometers (Additional file 1: ES XIII and ES I)
Results from computational approach 1900–1927 (Additional file 1: ES XIV)
Some exposure parameters from the period 1900–1910 and the corresponding ESAKa
KfiA at 1 m
Uncertainty estimate for a case with information on primary and secondary circuit
Beck (1904) 
Uncertainty estimate for a case with information on secondary circuit
Biddle (1905) 
After 1910 more of the published exposure data were complete and the computation of the ESAK was rather straightforward. Up to 1922 we assumed no filtration apart from the 0.85 mm glass and a column of air equal to the focus skin distance minus the radius of the X-ray tube (taken as 10 cm). After 1922 the use of filters was mentioned more frequently, and often recommended, but clearly not standard. The BSF was estimated casewise. The total number of computed doses before 1927 was 53.
Retrieved results from explicit (“modern”) dose measurements after 1927 (Additional file 1: ES XIV and ES XV)
After1927 ionization chamber based dose measurements were reported. A few doses had been measured with thermoluminescence dosemeters or a “KAP-meter”, the well-known large flat ionization chamber in the primary beam giving the product of kerma-in-air and beam area. In total 114 results were retrieved. Some additional points (n = 11) were still obtained by using our computational approach.
When starting the project, we hardly expected that dose reconstruction would be possible for the first decades of radiology. Surprisingly it was, with estimated uncertainties of the order of 40%–60% for the period 1900–1926. It appeared that these uncertainties were still surpassed by the variations in dose caused by differences in X-ray systems and their use. Large variations in dose persisted up to current times. Nevertheless, a substantial decrease in average dose was evident: since 1896 the skin dose for a radiograph in the pelvic region was roughly lowered by a factor of 400.
In our dose reconstruction we tried to use all available sources of information. Kotre et al.  reported the dose incurred by patients in the period 1899–1902, the single explicit dose estimate we found from before 1927. Chromoradiometer readings were available from three studies only, and although reportedly inaccurate, they are valuable because the results are methodologically independent from our computational approach. The latter was our mainstay for the period 1896–1926. For 1896–1899 an elementary assessment of the physics of heat loss by radiation of two 1896 X-ray tubes was at the basis of our computations. Due to lack of knowledge of the actual anode temperatures the results are rather speculative, but they seem compatible with the other data. The doses from later specifications of exposure parameters are more reliable, certainly after 1910, when all relevant physical parameters were evidently understood and experimentally assessable.
Our assessment of spark gap data, the various penetrometers, and the efficiency of different old röntgen systems regarding electrical transformation and X-ray production, is the first in its kind as far as we know. Simulations with SpekCalc were very helpful in extracting “modern” information from antiquated data. Generally our evaluations revealed considerable variability in all old data, but the relations between spark gap and high voltage were more concordant than the literature had led us to believe.
Comparison of measured doses (from the literature) with doses calculated using our computational approach was possible for the period 1927–1957, yielding a ratio of 0.97 ± 0.21. The relatively small standard deviation, compared to our pre-1927 uncertainty estimate of 40–60%, is probably due to more stable systems and increased accuracy of the data reported after 1927.
Nearly all low-lying points in Fig. 4 stem from investigators who showed that dose reduction was possible according to their standards. Braun  and Saupe  in 1928, Ardran and colleagues in 1938, 1939, 1953 and 1957 [43–45], Billings  in 1957 and Persliden  in 2002 belong to this group. Their doses are clearly not representative for the average practice at the time of reporting. But are other reported values? It is possible that X-ray users who published on radiation dose already adhered to stricter than average dose standards, implying that Fig. 4 might give a more favorable impression than genuine surveys would.
We saw only one other study with doses for an anteroposterior pelvic radiograph as a function of time: Huda et al.  published values for 1965–2002 that fall well within our cloud of data.
Because of the relatively large number (n = 182) of collected doses, their spread could quantitatively be assessed (Fig. 5). It turns out that the dose distributions are skewed and reminiscent of distributions reported in other dose surveys. The curve for the period 1896–1926 suggests that the spread in doses is largest in the first decades of radiology. This seems understandable considering the large variability in systems and their use. Some used already intensifying screens, which lowered the dose by a factor of about 5 to 10 compared to working without screens. Additional variability in the ion tube era was caused by the unintended change in gas pressure in the tube, whereas extremely high doses could result from inappropriate use. Mitchell et al. , Cassidy  and Deutschländer  reported wet desquamation or necrotization following exposure, implying doses higher than 18 Gy . The occurrence of such severe accidents suggests that the 1896–1926 distribution in Fig. 5 actually has a long tail to high doses.
For the time after 1927 we found explicit dose data, in the form of ESAK, KfiA and SAD. No systematic differences on the basis of data origin are apparent. Certainly in these later years the large spread in doses will have been caused by the users and their equipment, not by errors in measurements or calculations. Note that this variability signaled a large but unused potential for dose reduction by optimization. For some time now the realization of this potential is pursued by the introduction of diagnostic reference levels [53, 54]. This is for good reason, as our historical review proves it never happened by itself in the past century.
Reconstruction of the skin dose (ESAK) for pelvic radiography was performed for the period before 1927. For the time after 1927 doses were retrieved from the literature. At all times the doses showed a large spread. On average a decrease in dose with a factor of about 400 was observed. The methods of dose reconstruction developed here might be useful in future studies.
The authors thank Frank Jones, René van Tiggelen, Zahi Hakim, Rolf Behling and Cecile Jeukens for sharing information, and Madame Pallardy for permission to use Additional file 1: Figure ES4.
GJK conceived the study, performed the calculations and wrote the draft. GJK, PJK, JMAvE and KS collected the data and together with JEW wrote the final manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Kotre CJ, Little BG (2006) Patient and staff radiation doses from early radiological examinations (1899−1902). Br J Radiol 79:837–842. https://doi.org/10.1259/bjr/16982267 View ArticlePubMedGoogle Scholar
- (2005) 3 quantities and units for measurement and calculation in medical X-ray imaging. J ICRU 5:25–34. https://doi.org/10.1093/jicru/ndi025
- Wall BF (1996) How to assess the dose to the patient - refresher course R-02. International Radiation Protection Association, Vienna, pp 1–23Google Scholar
- Kaye GWC, Binks W (1929) The EVALUATION OF the X-ray pastille dose in Rontgen (r) units. Br J Radiol 2:553–578. https://doi.org/10.1259/0007-1285-2-24-553 View ArticleGoogle Scholar
- Poludniowski G, Landry G, DeBlois F, Evans PM, Verhaegen F (2009) SpekCalc: A program to calculate photon spectra from tungsten anode x-ray tubes. Phys Med Biol 54:N433–N438. https://doi.org/10.1088/0031-9155/54/19/N01
- Hubbell JH, Seltzer S (1995) Tables of X-ray mass attenuation coefficients and mass energy-absorption coefficients from 1 keV to 20 MeV for elements Z = 1 to 92 and 48 additional substances of dosimetric interest, NIST standard reference database 126. https://www.nist.gov/pml/x-ray-mass-attenuation-coefficients. Accessed: 9 Sept 2018
- XCOM NIST XCOM: Element/Compound/Mixture. https://physics.nist.gov/PhysRefData/Xcom/html/xcom1.html. Accessed 20 Aug 2018
- Graph Grabber 2.0 https://www.quintessa.org/software/downloads-and-demos/graph-grabber-2.0 Accessed: 1 Sept 2018
- Ullmann HJ (1921) The practical application of the sphere gap to roentgenotherapy. AJR Am J Roentgenol 8:195–199Google Scholar
- Surgeon General of the Army (1920) United States army X-ray manual. Paul B. Hoeber, New YorkGoogle Scholar
- AIEE (1969) E 61: Spark gap voltages, in CRC handbook of chemistry and physics. Chemical Rubber Company, ClevelandGoogle Scholar
- Spiegler G, Zakovsky J (1928) Der Halbwellenapparat der Röntgentechnik. Fortschritte auf dem Geb Röntgenstrahlen 37:718–728Google Scholar
- Peek FW (1915) Dielectric phenomena in high voltage engineering, 1st edn. McGraw-Hill Book Company, Inc., New YorkGoogle Scholar
- Schall WE (1932) X rays: Their origin, dosage and practical application, 4th edn. John Wright & Sons LTD, Stone Bridge, BristolGoogle Scholar
- Rieber F (1922) Standardization of the measurement of tube potential. AJR Am J Roentgenol 9:371–381Google Scholar
- Benoist ML (1902) Définition expérimentale des diverses sortes de rayons X, par le radiochromomètre. Comptes Rendus Acad Sci 134:225–227Google Scholar
- Hackenbruch PT, Berger W (1915) Mittlerer expositions-Tabelle in Milliampèresekunden. In: Vademekum für die Verwendung der Röntgenstrahlen und des Distraktionsklammerverfahren im Kriege. Otto Nemnich Verlag, Leipzig, p 111Google Scholar
- Luboshez BE (1925) On measuring and expressing X-ray quality in radiography. Br J Radiol BIR Sect 30:81–89. https://doi.org/10.1259/bir.1925.0019 View ArticleGoogle Scholar
- Kienböck R (1905) Über Dosimeter und das quantimetrische Verfahren. Fortschritte auf dem Geb Röntgenstrahlen 9:276–295Google Scholar
- Albers-Schönberg H, Walter B (1910) Die Röntgentechnik, 3rd edn. Lucas Gräfe & Sillem, HamburgGoogle Scholar
- Muir J, Reid A, Harlow FJ (1927) A manual of practical X-ray work. Paul B. Hoeber, Inc, New YorkGoogle Scholar
- Christen T (1914) Zur Theorie und Technik der Härtemessung. Fortschritte auf dem Geb Röntgenstrahlen 22:247–253Google Scholar
- Christen T (1913) Messung und Dosierung der Röntgenstrahlen. Fortschritte auf dem Geb Röntgenstrahlen Ergänzungsband 28:1–122Google Scholar
- Gaiffe G (1905) De l’emploi d’un milliampèremètre sur le circuit d’un tube de rayons X. Arch d‘électricité Médicale 13:61–63Google Scholar
- Coolidge WD (1913) A powerful Röntgen ray tube with a pure electron discharge. Phys Rev 2:409–430. https://doi.org/10.1103/PhysRev.2.409 View ArticleGoogle Scholar
- Goetze O (1923) Verfahren und Glühkathodenröntgenröhre zur Erzeugung scharfer Röntgenbilder (patent 370022)Google Scholar
- Harrison RM (1994) X-rays: Half-value layers 0.01-8.0 mm AI (approx. 6-150 kV peak potential difference). Br J Radiol Suppl 25:1–24Google Scholar
- National Council on Radiation Protection and Measurements (1981) Radiation protection in pediatric radiology: recommendations of the National Council on radiation protection and measurements. Report 68. NCRP, Washington, D.CGoogle Scholar
- Grigg ERN (1965) The trail of the invisible light. From X-Strahlen to radio(bio)logy. Charles C. Thomas, SpringfieldGoogle Scholar
- (1928) International X-ray unit of intensity. Br J Radiol 1:363–364. https://doi.org/10.1259/0007-1285-1-10-363
- Behnken H (1924) Die Eichung von Dosismessern in der Physikalisch-Technischen Reichsanstalt. Fortschritte auf dem Geb Röntgenstrahlen 31:479–482Google Scholar
- Gleßmer-Junike S (2015) X-Strahlen, radiometer und Hauteinheitsdosis. Thesis. Universität Hamburg. http://ediss.sub.uni-hamburg.de/volltexte/2015/7181/
- ICRU (1954) Recommendations of the international commission on radiological units. Br J Radiol 27(Suppl 6):243–245 doi.org/10.1259/0007-1285-27-316-243 Google Scholar
- ICRP (1955) Supplement no. 6. Recommendations of the ICRP. Br. J. Radiol. (Suppl. 6):100Google Scholar
- International Commission on Radiological Units and Measurements (ICRU) (1962) Radiation quantities and units - report 10a. National Bureau of standards handbook 84. U.S. Government Printing Office, WashingtonGoogle Scholar
- ICRU 33 A (1981) ICRU report 33 - radiation quantities and units pub: International commission on radiation units and measurements, Washington D.C. USA issued 15 April 1980, pp.25. J Label Compd Radiopharm 18:1398–1398. https://doi.org/10.1002/jlcr.2580180918 View ArticleGoogle Scholar
- Levy-Dorn M (1903) Schutzmassregelen gegen Röntgenstrahlen und ihre Dosirung. Dtsch Med Wochenschr 29:921–924View ArticleGoogle Scholar
- Dalton CHC (1927) The comparative efficiency of different types of transformers. Br J Radiol BIR Sect 32:443–446. https://doi.org/10.1259/bir.1927.0136 View ArticleGoogle Scholar
- Janus F (1909) Der Expositionsmesser, ein neues Hilfsinstrument für Röntgenaufnahmen. Fortschritte auf dem Geb Röntgenstrahlen 14:253–256Google Scholar
- Jaugeas F (1911) Rapid and instantaneous radiology. Arch Roentgen Ray 15:372–387. https://doi.org/10.1259/arr.1911.0130 View ArticleGoogle Scholar
- Braun R, Hase H, Küstner H (1928) Über die in der Diagnostik verabfolgten Dosen in R-Einheiten. Fortschritte auf dem Geb Röntgenstrahlen 38:385–398Google Scholar
- Saupe E (1928) Über die bei diagnostischen Arbeiten verabreichten Röntgenstrahlendosen in R-Einheiten. Fortschritte auf dem Geb Röntgenstrahlen 37:536–540Google Scholar
- Ardran GM, Crooks HE (1953) A comparison of radiographic techniques with special reference to dosage. Br J Radiol 26:352–357. https://doi.org/10.1259/0007-1285-26-307-352 View ArticlePubMedGoogle Scholar
- Ardran GM (1956) The dose to operator and patient in X-ray diagnostic procedures. Br J Radiol 29:266–269. https://doi.org/10.1259/0007-1285-29-341-266 View ArticlePubMedGoogle Scholar
- Ardran GM, Crooks HE (1957) Gonad radiation dose from diagnostic procedures. Br J Radiol 30:295–297. https://doi.org/10.1259/0007-1285-30-354-295 View ArticlePubMedGoogle Scholar
- Billings MS, Norman A, Greenfield MA (1957) Gonad dose during routine Roentgenography. Radiology 69:37–41. https://doi.org/10.1148/69.1.37 View ArticlePubMedGoogle Scholar
- Persliden J, Beckman KW, Geijer H, Andersson T (2002) Dose-image optimisation in digital radiology with a direct digital detector: an example applied to pelvic examinations. Eur Radiol 12:1584–1588. https://doi.org/10.1007/s00330-001-1206-0 View ArticlePubMedGoogle Scholar
- Huda W, Nickoloff EL, Boone JM (2008) Overview of patient dosimetry in diagnostic radiology in the USA for the past 50 years. Med Phys 35:5713–5728. https://doi.org/10.1118/1.3013604 View ArticlePubMedGoogle Scholar
- Mitchell JS, Haybittle JL (1955) Carcinoma of the skin appearing 49 years after a single diagnostic roentgen exposure. Acta Radiol 44:345–350View ArticleGoogle Scholar
- Cassidy P (1900) Report of a severe X-ray injury. Med Rec 57:180–181Google Scholar
- Deutschländer (1899) Beitrag zu dem Kapitel der Hautverbrennung durch Röntgenstrahlen. Fortschritte auf dem Geb Röntgenstrahlen 3:182–186Google Scholar
- Stewart FA, Akleyev AV, Hauer-Jensen M et al (2012) ICRP PUBLICATION 118: ICRP statement on tissue reactions and early and late effects of radiation in normal tissues and organs — threshold doses for tissue reactions in a radiation protection context. Ann ICRP 41:1–322. https://doi.org/10.1016/j.icrp.2012.02.001 View ArticlePubMedGoogle Scholar
- ICRP (2001) Diagnostic reference levels in medical imaging: Review and additional advice. ICRP supporting guidance 2. Ann ICRP 31:33–52Google Scholar
- Vañó E, Miller DL, Martin CJ et al (2017) ICRP Publication 135: Diagnostic Reference Levels in Medical Imaging. Ann ICRP 46:1–144. https://doi.org/10.1177/0146645317717209
- Snook HC (1908) A new Roentgen generator. Arch Roentgen Ray 13:186–188View ArticleGoogle Scholar
- Jennings WA (2007) Evolution over the past century of quantities and units in radiation dosimetry. J Radiol Prot 27:5–16. https://doi.org/10.1088/0952-4746/27/1/R01 View ArticlePubMedGoogle Scholar
- Mould RF (1993) A century of x-rays and radioactivity in medicine: with emphasis on photographic records of the early years. Institute of Physics Pub, BristolGoogle Scholar
- Macintyre J (1896) Roentgen rays: photography of renal calculus; description of an adjustable modification in the focus tube. Lancet 148:118View ArticleGoogle Scholar
- Rosenfeld G (1897) Die Diagnostik innerer Krankheiten mittels Röntgenstrahlen. Verlag von J.F. Bergman, WiesbadenGoogle Scholar
- Foveau de Courmelles F-V (1897) Traité de radiographie médicale et scientifique. Octave de Doin, ParisGoogle Scholar
- Gocht H (1898) Lehrbuch der Röntgen-Untersuchung: Zum Gebrauche für Mediciner. Verlag von Ferdinand Enke, StuttgartGoogle Scholar
- Isenthal AW, Snowden Ward H (1898) Practical radiography - a handbook of the applications of the X-rays, 2nd edn. Dawbarn & Ward, LondonGoogle Scholar
- Walsh D (1899) The Röntgen rays in medical work, 2nd edn. Baillière, Tindall and Cox, LondonGoogle Scholar
- Beck C (1904) Röntgen ray diagnosis and therapy. D. Appleton and company, New YorkView ArticleGoogle Scholar
- Holland CT (1904) Description of plate: Plate I: the left and right kidney regions: Plate a. (the left) shows the shadow of one stone. Plate B. (the right) shows the shadow of three stones. J Röntgen Soc 1:51–51. https://doi.org/10.1259/jrs.1904.0032 View ArticleGoogle Scholar
- Biddle JG (1905) Typical “roentgen” Equipments. James G. Biddle, Chestnut Street, PhiladelphiaGoogle Scholar
- Albers-Schönberg (1906) Die Röntgentechnik: Lehrbuch für Ärzte und Studierende, 2nd edn. Lucas Gräfe & Sillem, HamburgGoogle Scholar
- Arthur D, Muir J (1909) A manual of practical X-ray work. Rebman Company, New YorkView ArticleGoogle Scholar
- Béclère H (1910) Le Radiodiagnostic des affections du foie. G. Steinheil, Editeur, ParisGoogle Scholar
- Jaugeas F (1910) Radiographie rapide et radiographie instantanée. J Radiol Belge:601–629Google Scholar
- Tousey S (1910) Medical electricity and Röntgen rays. W.B. Saunders Company, PhiladelphiaGoogle Scholar