- Open Access
The future of hybrid imaging—part 2: PET/CT
© The Author(s) 2011
- Received: 1 October 2010
- Accepted: 13 January 2011
- Published: 20 February 2011
Since the 1990s, hybrid imaging by means of software and hardware image fusion alike allows the intrinsic combination of functional and anatomical image information. This review summarises the state-of-the-art of dual-modality imaging with a focus on clinical applications. We highlight selected areas for potential improvement of combined imaging technologies and new applications. In the second part, we briefly review the background of dual-modality PET/CT imaging, discuss its main applications and attempt to predict technological and methodological improvements of combined PET/CT imaging. After a decade of clinical evaluation, PET/CT will continue to have a significant impact on patient management, mainly in the area of oncological diseases. By adopting more innovative acquisition schemes and data processing PET/CT will become a fast and dose-efficient imaging method and an integral part of state-of-the-art clinical patient management.
- Hybrid imaging
The proposal to combine PET with CT was made in the early 1990s by Townsend, Nutt and co-workers. In addition to intrinsic image alignment, the anticipated benefit of a PET/CT hardware combination was to use the CT images to derive the PET attenuation correction factors . The first prototype PET/CT became operational in 1998 , designed and built by CTI PET Systems in Knoxville, Tenn., USA (now Siemens Healthcare) and clinically evaluated at the University of Pittsburgh. The design incorporated a single-slice spiral CT (Somatom AR.SP; Siemens Medical Solutions, Forchheim, Germany) and a rotating ECAT ART PET system (CTI PET Systems, Knoxville, Tennessee). Torso imaging using the prototype PET/CT took 1 h, or more. Over 300 cancer patients were imaged on the prototype. The results from the prototype demonstrated the importance of high resolution anatomy accurately co-registered to functional data [3–5]. This helped localise functional abnormalities and clarified equivocal situations, thus improving the accuracy and confidence of the data interpretation. The use of rapidly acquired, low-noise CT data in place of a lengthy conventional PET transmission acquisition reduced the overall duration of the examination .
Unknown to the developement team of the above prototype PET/CT, which served as a key impulse for the subsequent commercialisation of PET/CT development, another team at Gunma University in Japan had proposed a combination of a full-ring PET and CT as early as 1984. Their design was based on the two tomographs placed next to each other with a patient handling system traversing between the two units. However, this original work was not widely known, and was not further developed towards a commercial product.
PET/CT imaging has rapidly emerged as an important imaging tool in oncology . The success of PET/CT imaging is based on several factors. First, patients benefit from a comprehensive diagnostic anatomical and functional whole-body survey in a single session. Second, PET/CT provides more accurate diagnostic information than PET or CT alone. Third, PET/CT imaging allows radiation oncologists to use the functional information provided by PET for radiation treatment planning.
There is, at least for oncology, a growing body of literature that supports the accuracy of staging and restaging with PET/CT compared with either CT or PET acquired separately [8–10]. These improvements are incremental compared with PET, which alone demonstrates high levels of sensitivity and specificity for a wide range of disease states. Improved accuracy in primary diagnosis, staging and restaging has been documented for a variety of cancers, including head and neck, thyroid, lung, breast, oesophageal, colorectal, lymphoma, sarcoma, gastrointestinal stroma tumour (GIST), carcinoma of unknown primary, and melanoma [8, 11–13]. In summary, therefore, the improvement in accuracy of PET/CT compared with PET or CT alone for staging and restaging is statistically significant and averages 10–15% over all cancers .
An application on which PET/CT is also having an impact is that of radiotherapy treatment planning . In anticipation of application to radiotherapy treatment planning, the patient port on early PET/CT designs was increased to 70 cm in diameter. From the beginning, PET/CT provided more convenient and routine access to fused CT and PET images and early assessment of the consequences of using PET/CT in planning  was encouraging. As suggested, the convenience of having fused CT and PET images for every patient immediately following the PET/CT could not be matched by even the most sophisticated software. Increasingly, PET/CT systems are being acquired by radiation oncology departments and PET images are contributing directly to the definition of treatment volumes on CT-based plans [16, 17]. This will continue to be an area of growth, particularly in meeting the demands of intensity-modulated radiation therapy (IMRT) and the availability of radiation therapy devices able to treat tumours with surgical precision using sophisticated robotic techniques. Access to other PET biomarker mapping processes, such as hypoxia , that increase the radio-resistance of tumours could potentially expand the role of PET in radiation oncology. First PET/CT systems dedicated to PET-guided radiation therapy planning are on the market and, with a patient port diameter of 78–85 cm, facilitate the imaging of patients in the treatment position [19, 20].
As with radiation therapy planning, PET was already finding a role in assessing therapy response before the introduction of PET/CT. A detailed discussion of the use of PET/CT for monitoring therapy response may be found elsewhere . The combination of CT and PET for monitoring response offers a number of unique possibilities in spite of the technical difficulties associated with CT-based attenuation correction. First, the anatomical and functional volume of the tumour can be estimated, the former from CT measurements and the latter by summing all voxels with standardized uptake value (SUV)  above a threshold that defines malignancy. Therapy response can be assessed from changes in both these metrics or from a change in the total lesion glycolysis that is calculated as the product of the average SUV in the tumour and the volume .
Combined PET/CT gains importance for individualised treatment planning before radionuclide therapy, which is increasingly used as a treatment technique for a range of cancers, e.g. with radiolabelled peptides . To date, few treatments involve the use of dosimetry, either to plan treatment or to ascertain the delivered dose during treatment. Further, the correlation between the absorbed dose and the biological effect has been difficult to establish. PET is the most accurate imaging method for the determination of activity concentrations in vivo. PET imaging can be considered for pre-therapeutic treatment planning but ideally requires the use of a radioisotope from the same element as that used for treatment (e.g. 124I for 131I; 86Y for 90Y). For example, 124I-PET/CT dosimetry has emerged as a valuable tool for confirming or revising staging and for planning therapy in differentiated thyroid cancer [25–29]. Some groups proposed to routinely perform this procedure before the first radioiodine therapy (RIT) with 131I for thyroid remnant ablation in differentiated thyroid cancer (DTC) patients at high risk of persistent disease, i.e., patients with tumour invasion of the perithyroidal soft tissue, histologically proven lymphatic involvement, distant metastases, unfavourable histotypes, or age ≤18 years .
The rationale is to detect patients with previously occult loco-regional or distant metastases who may benefit from RIT with an individually tailored escalated activity. These patients receive such RIT when lesion uptake in the dosimetry procedure permits reasonably high confidence that the RIT will yield tumour doses exceeding 100 Gy, while staying within established limits for avoiding bone marrow toxicity. Patients without sufficient tumour uptake in the dosimetry procedure still receive RIT with a standard activity for remnant ablation according to national guidelines. In patients given RIT for metastatic involvement, 124I-PET/CT dosimetry is repeated before subsequent courses of RIT. This concept can serve as a matrix for other radionuclide therapies .
Since the development in CT has been driven primarily by cardiology demands for faster acquisitions, the top-of-the-line PET/CT now incorporates 128-slice CT and is in principle ideally suited to cardiac PET/CT. The ability to image both structure and function could, for example, enhance the characterisation of atherosclerotic plaques by imaging the inflammatory process associated with the plaque. The combination of CT angiography together with a measurement of myocardial perfusion using a PET tracer such as 82Rb-rubidium or 13N-ammonia could, in a single examination, assess both the integrity of the cardiac arteries and the metabolic consequences for the myocardium .
Cardiac PET/CT applications are still in their infancy and have recently encountered a number of difficulties. Obviously, the effects of cardiac and respiratory motion are critical for these studies. The problems of mismatch associated with CT-based attenuation correction are potentially more serious for cardiac studies than they are for oncology, in that all cardiac studies will be affected rather than just those whole-body studies with lesions in certain sensitive regions such as the lung. This mis-registration results in what appears to be perfusion deficits in segments of the heart associated with the misalignment.
A number of different strategies are being developed to address this issue, including (1) manual realignment of CT and PET, (2) acquiring a cine CT of the breathing motion and generating an average CT for attenuation correction and (3) acquiring multiple CT data sets to ensure at least one matches the PET image as closely as possible. Obviously, the role of PET/CT in cardiology has yet to be determined, but if a strong clinical demand exists it is to be expected that transient technical challenges such as the misalignment issue will ultimately be resolved.
The main objective of PET is to detect metabolic abnormalities, synonymous with disease (or responses to functional tests) and to quantify these changes in metabolic uptake patterns with respect to normal background activities. Therefore, the diagnostic accuracy of PET is not only dependent on spatial resolution but, more so, on the sensitivity of the detection system. PET sensitivity describes the ability of the PET system to utilise a fraction of the injected activity for signal detection. The higher the sensitivity, the more signal is detected for a given activity injected, thereby providing increased detectability of lesions.
PET is intrinsically a three-dimensional (3D) imaging methodology, replacing physical collimation required for single-photon imaging with the electronic collimation of coincidence detection. The availability of PET systems with retractable inter-plane septa encouraged the use of 3D methodology, at least for the brain, where the net increase of a factor of 5 in sensitivity could be realised . The situation for whole-body imaging is far less favourable, in part due to the presence of significant activity just outside the imaging field of view (FOV) in most bed positions. With the appearance of LSO (lutetium oxyorthosilicate)- and GSO (gadolinium silicate)-based PET, which could be operated with short coincidence time windows (4.5–6 ns) and higher lower-energy thresholds (400–450 keV) compared with 10–12 ns and 350 keV for a typical BGO-based PET, significantly improved whole-body image quality has been achieved in 3D .
Despite the increase in volume sensitivity from increased axial coverage (Fig. 2), 3D PET suffers from a sharp decrease in sensitivity at both ends of the axial FOV. This drop in sensitivity mandates slightly overlapping bed positions when imaging larger imaging fields, such as the torso or the whole-body. Sensitivity in the overlapping regions is slightly lower than the sensitivity in the centre of the axial FOV. Dahlbom and co-workers suggested as early as 1992 a continuous axial sampling mode by moving the patient bed in steps equal to half the plane separation . As a result, better image uniformity and sensitivity was achieved, together with a reduction in noise compared with standard multi-bed acquisition modes. Further refinements of the continuous acquisition mode for PET were suggested by the same authors . However, the improvements in clinical image quality were less obvious than in simulation studies. Nonetheless, continuous bed motion based on modern hardware and highly accurate bed positioning and position tracing should yield a significant improvement in whole-body PET image quality. Continuous PET acquisition would then be complementary to continuous, spiral CT acquisitions as performed in PET/CT. Finally, continuous bed motion supports the free definition of co-axial imaging ranges that otherwise would be limited by an integer number of PET bed positions, and, thus, lead to an overexposure of the patients in the lower part of the co-axial imaging range.
Co-registered CT images may also be used to improve partial volume correction by dividing the standardised uptake value (SUV) from the PET image with a recovery coefficient based on the spherical tumour diameter. As tumours generally have a complex shape, a more sophisticated partial volume correction method is desirable . Thus, for both technical and practical reasons, PET/CT is continuing to successfully promote the use of PET for monitoring response to different forms of therapy.
The PET systems with fast scintillators and electronics can measure this time difference within a certain resolution. For example, for a PET system with a coincidence timing resolution of 500 ps, the spatial uncertainty on the position of the annihilation is 7.5 cm. For a 40-cm diameter uniform distribution and a 7.5-cm uncertainty, the increase in SNR is a factor of about 2.3. As the TOF resolution improves, the spatial uncertainty decreases and the SNR increases by a larger factor. TOF PET systems must demonstrate good timing resolution that is stable over time so as to avoid frequent detector recalibration . While promising, the clinical impact of TOF PET has yet to be established [46, 47].
There has been significant progress during the past few years in image reconstruction methods through the introduction of statistically based algorithms into the clinical setting. Iterative reconstruction methods became of clinical interest in nuclear medicine imaging with the accelerated convergence achieved by the ordered subset expectation maximisation (OSEM) algorithm . Lately iterative image reconstruction has been proposed for clinical CT imaging as well, particularly as a method of keeping radiation exposure at low levels for an overall reduced patient exposure .
Further improvement can be achieved by eliminating the rebinning step and implementing OSEM fully in 3D with corrections for randoms, scatter, attenuation and detector efficiency variations incorporated into the system model . Finally, in a recent development termed high-definition (HD) PET, the detector spatial response function has also been included in the reconstruction model . The point spread function varies throughout the FOV owing to the oblique penetration of the detectors by annihilation photons. By measuring this variability and then modelling the PSF, improved and near-uniform spatial resolution can be achieved throughout the FOV (Fig. 4).
All vendors provide comparable software capable of producing clinical images of high quality . The greatest outstanding effect on image quality and a challenge to reconstruction algorithms is now the size of the patient, a significant problem given the current levels of obesity in industrialised countries.
However, some image artefacts must be accepted as inherent to PET/CT (and SPECT/CT) imaging. Of particular importance in the assessment of head and neck cancer is the presence of dental fillings  (Fig. 7b). A number of metal artefact reduction techniques have been explored in research [57, 58] but still await routine implementation in PET/CT.
Nonetheless, after a decade of clinical evaluation PET/CT continues to have a significant impact on patient management, mainly in the area of oncological diseases. Image artefacts inherent to PET/CT can be detected, interpreted and—in many cases—corrected or avoided by well-trained users. By adopting more innovative acquisition schemes and data processing PET/CT will become a fast and dose-efficient imaging method and manifest itself as an integral part of state-of-the-art patient management.
We are indebted to Dale Bailey (Sydney), Andreas Bockisch (Essen), Claude Comtat (CEA Orsay), Bernd Pichler (Tübingen), York Hämisch (Bioscan), Matthias Hofmann (Tübingen), Ora Israel (Haifa), Antonis Kalemis (Philips London), Armin Kolb (Tübingen), Paul E Kinahan (Seattle), Thomas Krause (Bern), Roger Lecomte (Sherbrooke), Marcus Lonsdale (Copenhagen), Bernd Schweizer (Philips Research Aachen), Rainer Veigel (Philips Zurich) for helpful discussions, advice and support materials.
Conflicts of interest
T.B. is the founder and president of Switzerland-based cmi-experts GmbH.
D.T. acts as scientific advisor to cmi-experts of Zurich, Switzerland and RefleXion Medical of Stanford, Calif., USA. He received royalty payments from Siemens Healthcare related to the invention of the PET/CT.
J.C. is co-founder of Momentum Biosciences and Sofie Biosciences, Los Angeles, Calif., USA and serves as an advisor to cmi-experts GmbH, Switzerland.
L.S.F. is an associate of ZRN Grevenbroich and Dormagen, and serves as an advisor to cmi-experts of Zurich, Switzerland. He has received speaker fees from Siemens, Philips and Genzyme.
- Kinahan PE et al (1998) Attenuation correction for a combined 3D PET/CT scanner. Med Phys 25(10):2046–2053View ArticlePubMedGoogle Scholar
- Beyer T et al (2000) A combined PET/CT tomograph for clinical oncology. J Nucl Med 41(8):1369–1379PubMedGoogle Scholar
- Charron M et al (2000) Image analysis in patients with cancer studied with a combined PET and CT scanner. Clin Nucl Med 25(11):905–910View ArticlePubMedGoogle Scholar
- Kluetz PG et al (2000) Combined PET/CT imaging in oncology: Impact on patient management. Clin Positron Imaging 3(3):1–8Google Scholar
- Meltzer C, Martinelli M, Beyer T (2001) Whole-body FDG PET imaging in the abdomen: value of combined PET/CT. J Nucl Med 42:35PGoogle Scholar
- Kinahan P, Hasegawa B, Beyer T (2003) X-ray-based attenuation correction for PET/CT scanners. Semin Nucl Med 33(3):166–179View ArticlePubMedGoogle Scholar
- Townsend D (2008) Multimodality imaging of structure and function. Phys Med Biol 53(4):R1–R39View ArticlePubMedGoogle Scholar
- Czernin J, Allen-Auerbach M, Schelbert H (2007) Improvements in cancer staging with PET/CT: literature-based evidence as of September 2006. J Nucl Med 48 (Suppl 1):78S–88SPubMedGoogle Scholar
- Hellwig D, Baum R, Kirsch C (2009) FDG-PET, PET/CT and conventional nuclear medicine procedures in the evaluation of lung cancer: a systematic review. Nuklearmedizin 48(2):59–69PubMedGoogle Scholar
- Ben-Haim S, Ell P (2009) 18 F-FDG PET and PET/CT in the evaluation of cancer treatment response. J Nucl Med 50(1):88–99View ArticlePubMedGoogle Scholar
- Facey K et al (2007) Overview of the clinical effectiveness of positron emission tomography imaging in selected cancers. Health Technol Assess 11(44):iii–iv, xi-267View ArticlePubMedGoogle Scholar
- Poeppel T et al (2009) PET/CT for the staging and follow-up of patients with malignancies. Eur J Radiol 70(3):382–392View ArticlePubMedGoogle Scholar
- Weber W, Grosu A, Czernin J (2008) Technology Insight: advances in molecular imaging and an appraisal of PET/CT scanning. Nat Clin Pract Oncol 5(3):160–170View ArticlePubMedGoogle Scholar
- Ford E et al (2009) 18 F-FDG PET/CT for image-guided and intensity-modulated radiotherapy. J Nucl Med 50(10):1655–1665PubMed CentralView ArticlePubMedGoogle Scholar
- Grosu A-L et al (2005) Positron emission tomography for radiation treatment planning. Strahlenther Onkol 8:483–499View ArticleGoogle Scholar
- Lecchi M et al (2008) Current concepts on imaging in radiotherapy. Eur J Nucl Med Mol Imaging 35(4):821–837View ArticlePubMedGoogle Scholar
- Dirix P et al (2009) Dose painting in radiotherapy for head and neck squamous cell carcinoma: value of repeated runctional imaging with 18F-FDG PET, 18F-fluoromisonidazole PET, diffusion-weighted MRI, and dynamic contrast-enhanced MRI. J Nucl Med 50(7):1020–1027View ArticlePubMedGoogle Scholar
- Dehdashti F et al (2003) In vivo assessment of tumor hypoxia in lung cancer with 60Cu-ATSM. Eur J Nucl Med Mol Imaging 30(6):844–850View ArticlePubMedGoogle Scholar
- Lonsdale M, Beyer T (2010) Dual-modality PET/CT instrumentation-today and tomorrow. Eur J Radiol 73(3):452–460View ArticlePubMedGoogle Scholar
- Sattler B et al (2010) PET/CT (and CT) instrumentation, image reconstruction and data transfer for radiotherapy planning. Radiother Oncol 96:288–297View ArticlePubMedGoogle Scholar
- Weber W, Figlin R (2007) Monitoring cancer treatment with PET/CT: does it make a difference? J Nucl Med 48(Suppl 1):36S–44SPubMedGoogle Scholar
- Thie J (2004) Understanding the standardized uptake value, its methods, and implications for use. J Nucl Med 45(5):1431–1434PubMedGoogle Scholar
- Larson SM et al (1999) Tumor treatment response based on visual and quantitative changes in global tumor glycolysis using PET-FDG imaging: The visual response score and the change in total lesion glycolysis. Clin Positron Imaging 2(3):159–171View ArticlePubMedGoogle Scholar
- Cremonesi M et al (2006) Dosimetry in peptide radionuclide receptor therapy: a review. J Nucl Med 47:1467–1475PubMedGoogle Scholar
- Frey P, Townsend D, Flattet A (1986) Tomographic imaging of the human thyroid using I-124. J Clin Endocrinol Metab 63:918–927View ArticlePubMedGoogle Scholar
- Pentlow K, Graham M, Lambrecht R (2007) Quantitative imaging of iodine-124 with PET. J Nucl Med 37:1557–1562Google Scholar
- Erdi YE et al (1999) Radiation dose assessment for I-131 therapy of thyroid cancer using I-124 PET imaging. Clin Positron Imaging 2:41–46View ArticlePubMedGoogle Scholar
- Eschmann SM et al (2002) Evaluation of dosimetry of radioidodine therapy in benign and malignant thyroid disorders by means of iodine-124 and PET. Eur J Nucl Med 29(6):760–767View ArticleGoogle Scholar
- Jentzen W et al (2008) Optimized 124I PET dosimetry protocol for radioiodine therapy of differentiated thyroid cancer. J Nucl Med 49(6):1017–1023View ArticlePubMedGoogle Scholar
- Freudenberg L et al (2010) Lesion dose in differentiated thyroid carcinoma metastases after rhTSH or thyroid hormone withdrawal: (124)I PET/CT dosimetric comparisons. Eur J Nucl Med Mol Imaging 37(12):2267–2276View ArticlePubMedGoogle Scholar
- Bockisch A et al (2006) 124I in PET imaging: impact on quantification, radiopharmaceutical development and distribution. Eur J Nucl Med 33(11):1247–1248View ArticleGoogle Scholar
- Menezes L et al (2009) Assessment of left ventricular function at rest using rubidium-82 myocardial perfusion PET: comparison of four software algorithms with simultaneous 64-slice coronary CT angiography. Nucl Med Commun 30(12):918–925View ArticlePubMedGoogle Scholar
- Townsend DW (2004) From 3-D positron emission tomography to 3-D positron emission tomography/computed tomography: what did we learn? Mol Imaging Biol 6(5):275–290View ArticlePubMedGoogle Scholar
- Watson CC et al (2005) Optimizing injected dose in clinical PET by accurately modeling the counting-rate response functions specific to individual patient scans. J Nucl Med 46(11):1825–1834PubMedGoogle Scholar
- Dahlbom M et al (1992) Methods for improving image quality in whole body PET scanning. IEEE Trans Nucl Sci 39(4):1079–1083View ArticleGoogle Scholar
- Dahlbom M, Reed J, Young J (2001) Implementation of true continuous bed motion in 2-D and 3-D whole-body PET scanning. IEEE Trans Nucl Sci 48(4):1465–1469View ArticleGoogle Scholar
- Cheng G et al (2010) Current status of sentinel lymph-node biopsy in patients with breast cancer. Eur J Nucl Med Mol Imaging. doi:10.1007/s00259-010-1577-z Google Scholar
- Freudenberg L et al (2010) PET versus PET/CT dual-modality imaging in evaluation of lung cancer. Thorac Surg Clin 20(1):25–30View ArticlePubMedGoogle Scholar
- Panin V et al (2006) Fully 3-D PET reconstruction with system matrix derived from point source measurements. IEEE Trans Med Imaging 25(7):907–921View ArticlePubMedGoogle Scholar
- Soret M, Bacharach S, Buvat I (2007) Partial-volume effect in PET tumor imaging. J Nucl Med 48(6):932–945View ArticlePubMedGoogle Scholar
- Budinger T (1983) Time-of-flight positron emission tomography: status relative to conventional PET. J Nucl Med 24(1):73–78PubMedGoogle Scholar
- Moses W (2003) Time of flight in PET revisited. IEEE Trans Nucl Sci 50(5):1325–1330View ArticleGoogle Scholar
- Surti S et al (2007) Performance of Philips Gemini TF PET/CT scanner with special consideration for its time-of-flight imaging capabilities. J Nucl Med 48(3):471–480PubMedGoogle Scholar
- Surti S, Karp J (2009) Experimental evaluation of a simple lesion detection task with time-of-flight PET. Phys Med Biol 54(2):373–384PubMed CentralView ArticlePubMedGoogle Scholar
- Daube-Witherspoon M et al (2006) Influence of the time-of-flight kernel accuracy in TOF-PET reconstruction. IEEE MIC. doi:10.1109/NSSMIC.2006.354230 Google Scholar
- Murray I et al (2010) Time-of-flight PET/CT using low-activity protocols: potential implications for cancer therapy monitoring. Eur J Nuc Med Mol I 37(9):1643–1653View ArticleGoogle Scholar
- Lois C et al (2010) An assessment of the impact of incorporating time-of-flight information into clinical PET/CT imaging. J Nucl Med 51(2):237–245PubMed CentralView ArticlePubMedGoogle Scholar
- Hudson HM, Larkin RS (1994) Accelerated image reconstruction using ordered subsets of projection data. IEEE Trans Med Imaging 13:601–609View ArticlePubMedGoogle Scholar
- Flohr T et al (2010) Pushing the envelope: new computed tomography techniques for cardiothoracic imaging. J Thorac Imaging 25(2):100–111View ArticlePubMedGoogle Scholar
- Comtat C et al (2002) Clinically feasible reconstruction of 3D whole-body PET/CT data using blurred anatomical labels. Phys Med Biol 47:1–20View ArticlePubMedGoogle Scholar
- Kalender WA et al (1990) Spiral volumetric CT with single-breath-hold technique, continuous transport, and continuous scanner rotation. Radiology 176:181–183View ArticlePubMedGoogle Scholar
- Burger C et al (2002) PET attenuation coefficients from CT images: experimental evaluation of the transformation of CT into PET 511-keV attenuation coefficients. Eur J Nucl Med 29(7):922–927View ArticleGoogle Scholar
- Nakamoto Y et al (2003) Accuracy of image fusion of normal upper abdominal organs visualized with PET/CT. Eur J Nucl Med Mol I 30(4):597–602View ArticleGoogle Scholar
- Beyer T et al (2004) Acquisition protocol considerations for combined PET/CT imaging. J Nucl Med 45(Suppl 1):25S–35SPubMedGoogle Scholar
- Beyer T, Veit-Haibach P (2006) Oncology tasks require disease-specific PET/CT. Diagn Imaging Europe December 2006/January 2007:14-18Google Scholar
- Kamel EM et al (2003) Impact of metallic dental implants on CT-based attenuation correction in a combined PET/CT scanner. Eur Radiol 13:724–728View ArticlePubMedGoogle Scholar
- Schäfers K, Raupach R, Beyer T (2006) Combined 18 F-FDG-PET/CT imaging of the head and neck. An approach to metal artifact correction. Nuklearmedizin 45:219–222PubMedGoogle Scholar
- Hamill J et al (2006) A knowledge-based method for reducing attenuation artefacts caused by cardiac appliances in myocardial PET/CT. Phys Med Biol 51(11):2901–2918View ArticlePubMedGoogle Scholar
Open Access This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.