- Pictorial Review
- Open Access
Breast magnetic resonance imaging: tips for the diagnosis of silicone-induced granuloma of a breast implant capsule (SIGBIC)
Insights into Imaging volume 8, pages439–446(2017)
Complications resulting from the placement of silicone breast implants are becoming more frequent in our clinical practice. This is due to the increase in breast aesthetic surgeries at the beginning of the century, where breast augmentation using silicone implants was the main intervention performed. Generally, studies that discuss the complications of breast implants are restricted to reports of intra- or extra-capsular ruptures, contractures and haematomas. Currently, much importance has been given to anaplastic large cell lymphoma (ALCL) as a more severe complication related to silicone implants. Recently, granuloma formation induced by silicone particle bleeding from intact breast implants has been described when the free silicone comes into contact with the fibrous capsule of the implant. Few studies have demonstrated the characteristics and diagnostic keys for this entity. The objective of this study is to present cases of SIGBIC diagnosed in our service and to discuss the main findings that allow its diagnosis.
• Breast implants induce fibrous capsule formation at the periphery of the implant.
• Gel bleeding is inherent in all types of silicone breast implants.
• Gel bleeding induces silicone-induced granuloma of breast implants.
• Main diagnostic tips: heterogeneous mass, black-drop sign and late enhancement.
In the early 2000s, an epidemic of breast augmentation surgeries was observed in Brazil. The economic boom experienced at the beginning of the century associated with socio-cultural factors led to this epidemic. Seventeen years later, some repercussions of these interventions are being observed in our society.
According to the International Society of Aesthetic Surgery (ISAP), Brazil has the second highest number of aesthetic surgeries worldwide, second only to the USA. It is estimated that in 2015, 1.22 million aesthetic surgeries were performed. Of these, breast augmentation with the placement of breast implants was the second most performed surgery, the first being liposuction. In that year approximately 158,950 breast implants were placed in Brazilian women, the majority for aesthetic purposes. Silicone implants are the most used in our society .
The medical literature describes many complications related to the presence of silicone implants, the main ones being seromas, infections, haematomas, and intra- and extracapsular ruptures [2,3,4]. Anaplastic large cell lymphoma (ALCL) is also highlighted as a more severe complication. In the past year, the Federal Food and Drug Administration (FDA) of the USA has been concerned about the increased incidence of lethal cases of ALCL .
Recently, findings of silicone-induced granuloma have been reported in the breast implant capsule, termed silicone-induced granuloma of the breast implant capsule (SIGBIC) . This finding was reported in a patient with intact silicone breast implants with clinical findings of capsular contracture submitted to diagnostic breast magnetic resonance imaging (MRI).
The purpose of this pictorial essay is to describe the mechanism of development of this granulomatous process and to demonstrate the main findings by magnetic resonance imaging. The findings are based on diagnostic breast MRI scans performed at our service. Informed consent was obtained for all patients. Approval by the institutional ethics committee was waived because of the nature of the manuscript.
Pathophysiology of SIGBIC
To understand the formation of GISGBIC, it is important to know the pathophysiology involved in its formation process. Some concepts are of extreme importance for understanding this process, such as knowing the elements that participate in granuloma formation, the phenomenon of gel bleeding in intact breast implants and autoimmune reactions to silicone-content particles.
Placement of the breast implant, a non-biological material, in the breast induces fibrous capsule formation at the periphery of the implant. The cell composition of the capsule has been extensively studied, and the results point to a role of the immune system in the pathologies developed in the capsule. The predominant cell types reported within the fibrous capsule are macrophages, lymphocytes and fibroblasts. Mast cells are also reported, which can express renin, histamine and tumour growth factor β-1 (TGF-β). It is supposed that the mast cells activate fibroblasts, which produce collagen, one of the mechanisms of capsule contracture. A role of T cells has also been hypothesised. The fibrous capsule works as a physical barrier that isolates the intra-capsular contents from the rest of the body .
The phenomenon of gel bleeding is inherent in all types or models of silicone breast implants, which are soft, round or cohesive with anatomical shape. Even the most modern implants that contain an extra outer layer of protection do not prevent silicone bleeding; it only slows down the process. The phenomenon of gel bleeding determines the weakening of the elastomer independent of the brand. This is supposed be the triggering factor for implant rupture .
There is a virtual space between the breast implant and the fibrous capsule, where theoretically there should be no structure. Figure 1 schematically shows the granuloma formation induced by silicone bleeding from an intact breast implant in the fibrous capsule (Fig. 1) .
Tervaert and Kappel  described the association between autoimmunity and silicone exposure as follows:
Contact of silicone-content particles bleeding from the intact breast implant with the fibrous capsule;
Capture of these silicone-content particles by macrophages with entrapment within the lysosomes, activating the macrophages;
Macrophage apoptosis by the production of cytokines [interleukin-1b, reactive oxygen species (ROS) and reactive nitrogen species]. It releases silicone-content particles, which are captured by another macrophage, generating a vicious circle;
Massive production of interleukin-17 with neutrophil influx activation and ROS production;
Myeloid granular enzyme (myeloperoxidase) release, which produces hypohalous acids; these have antimicrobial activity;
There is also lymphocyte stimulation by a type 2 inflammatory response, increasing the levels of IgE and IgG1 associated with chronic activation of T lymphocytes. This chronic activation is probably due to the dysfunction of negative regulation and T cells;
As there is a physical barrier between the rest of the body and the intracapsular space, the vascularisation inside the capsule is poor, which makes the process indolent.
Clinical findings of SIGBIC
In 2015, Tervaert and Kappel reported that of the 600 patients evaluated in an ambulatory immunology clinic, 32 had silicone breast implants. All patients with silicone implants presented with silicone implant incompatibility syndrome (SIIS), whereas none of the patients with saline or mixed implants presented signs of autoimmune disease. The main symptoms associated with SIIS were capsular contracture and/or systemic manifestations such as fatigue, arthralgia, myalgia, asthenia and/or fever. Of the 32 patients, a systemic autoimmune disease was diagnosed in half of them. These patients might have had undiagnosed SIGBIC .
In this pictorial essay, all patients with SIGBIC presented with clinical signs of implant capsular contracture, characterised by hardening and pain in the compromised breast. These symptoms are usually interspersed by periods of spontaneous remission or by use of anti-inflammatory/corticosteroid therapy as reported by the patients. These findings suggest a relationship between silicone implants and autoimmune disease.
The time from implant placement surgery ranged from 2 to 20 years. The age of the patients ranged from 30 to 71 years. None of the patients had other chronic diseases.
Of the eight patients described in the study, three have had a history of repair surgery after breast cancer treatment and the other five of breast augmentation (aesthetic surgery). Six patients reported pruritus in the upper and/or lower extremities with beginning symptoms related to implant placement.
Breast magnetic resonance imaging for breast implant evaluation
Magnetic resonance is the examination of choice for evaluating breast implants. Currently, it is mainly used to evaluate intra- or extra-capsular ruptures of breast implants. Usually a specific protocol is used for the evaluation of breast implants, with dedicated sequences that perform water and silicone suppression. Many services choose not to perform intravenous contrast for evaluation of implants. The absence of contrast media makes it impossible to distinguish SIGBIC from intra-capsular haematoma. Searching the bibliographic databases online on the internet, only a few references to this entity were found.
Our hospital is a regional reference centre for breast cancer diagnostics and treatment. Breast MRI scans correspond to about 15% of the MRI scans performed at our institution. Approximately 38% of the breast MRI indications at our service are for breast cancer screening in high-risk patients, while the other 62% are diagnostic scans as established by the Colégio Brasileiro de Radiologia (CBR): evaluation of breast implants, staging of breast carcinoma and/or evaluation of response to neoadjuvant therapy, and occult carcinoma screening .
Until 2015, the breast implant MRI protocol at our institution was performed exclusively without contrast media. We only added water and silicone saturation sequences to the standard protocol. However, in the cancer screening patient group with breast implants who underwent contrast dynamic sequences and who presented intracapsular seroma, we observed that some had a solid lesion instead of seroma, with late contrast enhancement. From this finding, we started to adopt the dynamic series for all patients with breast implants to our protocol.
Currently, about 32% of our patients who performed breast MRI scans have breast implants. Of them, approximately 56% have capsular contracture, and 12% have findings compatible with SIGBIC.
Imaging findings of SIGBIC
Based on the cases of our clinical practice, we describe the tips to facilitate the diagnosis of SIGBIC. Among the MRI findings, we highlight an intra-capsular mass, black drop sign and late contrast enhancement. As associated findings, we note intra-capsular collection and capsular contracture (Table 1).
An intracapsular mass usually has a high heterogeneous signal in T2-weighted sequences and hyposignal in T1-weighted sequences. It is a slow-growing mass that has a compressive effect on the breast implant. Its main differential diagnosis is intracapsular haematoma, which makes the use of intravenous contrast essential (Figs. 2, 3, 4 and 5).
Black drop sign
This sign was found in all of our studied cases. It consists of a marked hyposignal focus at the interface between the breast implant elastomer and the granuloma on the dynamic sequence. This sign was visualised in the area of granuloma formation in all our cases (Fig. 6).
Late contrast enhancement
The contrast enhancement of the mass is impaired by the presence of the barrier exerted by the intact fibrous capsule. Mass enhancement is progressive, with a type I dynamic curve pattern. Usually nodular areas of greater vascularisation appear within the mass. Without the late phase sequences after the use of contrast media, at least 4 min, differentiation from intracapsular haematoma is difficult (Fig. 7).
Intracapsular haematoma/seroma may be associated with the mass. It is characterised by fluid collection that does not enhance with the contrast media (Fig. 7). The capsular contracture is determined by thickening of the fibrous capsule associated with an increase of the anteroposterior diameter. Enhancement of the fibrous capsule may be present to different degrees (Fig. 8).
Imaging of granuloma in patients with intracapsular implant rupture
For comparative purposes, we describe a case of intracapsular breast implant rupture associated with granuloma of the fibrous capsule.
The signs and symptoms are very similar to those found in silicone bleeding; however, the integrity of the implant elastomer is not maintained. The enhancement pattern and morphology of the granuloma are identical to those found when associated with intact implants since these patients may present the SIIS (Fig. 9).
SIGBIC after implant removal and capsulectomy
The treatment of SIGBIC consists of implant removal and capsulectomy. If there is a remnant of the fibrous capsule in the implant site, fluid collection can be observed at the surgical site. It is recommended not to replace the silicone implant with another with the same content (Fig. 10).
It is important to emphasise that percutaneous diagnostic biopsies should be avoided in these cases. Performing the biopsy breaks the fibrous capsule barrier and exposes the rest of the body to the SIGBIC.
Preliminarily, we observed that some patients who undergo breast implant replacement because of capsule contracture and who in retrospective analysis have signals of SIGBIC at the pre-operative MRI scan tend to present early post-operative complications such as capsule contracture and intracapsular seroma.
We are carrying out a prospective study to evaluate the relationship between early contractures after implant replacement and SIGBIC.
Patient management and follow-up
The medical management adopted for the SIGBIC (surgery or follow-up) in this pictorial essay was determined by the mastologists responsible for each patient.
Of the eight patients described, six chose surgical removal of the implant with capsulectomy. At histology, all six of these patients presented foreign body granuloma (silicone-content particles) with histological elements similar to those described by Cohen et al.
In the surgical specimen, SIGBIC presented as focal hardened areas at the fibrous capsule in the topography corresponding to the MRI findings. None of the silicone implants showed signs of rupture.
Two patients opted to have MRI scan follow-up despite being informed of the presence of SIGBIC. Despite the recommendation of implant removal and capsulectomy, half-yearly follow-up was suggested for these two cases.
As described, it is very important for the radiologist to recognise the characteristics of the SIGBIC to perform early diagnosis of this pathology. The diagnosis is relatively easy; however it requires using contrast medium where the lesions will be better evidenced in the late dynamic phases. We speculate that SIGBIC, when it overcomes the fibrous capsule, may determine a systemic autoimmune reaction that triggers a manifestation similar to anaplastic large cell lymphoma. Further studies should be performed to determine the association between SIGBIC and ALCL.
Silicone-induced breast granuloma
Anaplastic large cell lymphoma
International Society of Aesthetic Surgery
Federal Drug Administration
Magnetic resonance imaging
- TGF- β:
Tumour growth factor β-1
Reactive of oxygen species
Colégio Brasileiro de Radiologia
isaps.org [homepage on the internet]. Hanover: International Society of Aesthetic Plastic Surgery. Available from: https://www.isaps.org/Media/Default/global-statistics/2016%20ISAPS%20Results.pdf. Accessed 10 May 2017
Peters W, Fornasier V, Howarth D (2014) Late unilateral hematoma after breast augmentation. Plast Surg (Oakv) 22(1):18–21
Rubino C, Brongo S, Pagliara D et al (2014) Infections in breast implants: a review with a focus on developing countries. J Infect Dev Ctries 8(9):1089–1095
Spear SL, Rottman SJ, Glicksman C, Bown M, Al-Attar A (2012) Late seromas after breast implants: theory and practice. Plast Reconstr Surg 30(2):423–435
fda.gov [homepage on the internet]. Silver Spring: US Food and Drug Administration; [Updated August 13, 2013]. Available from: http://www.fda.gov/MedicalDevices/ProductsandMedicalProcedures/ImplantsandProsthetics/BreastImplants/ucm064348.htm. Accessed 10 May 2017
Fleury EF, Rego MM, Ramalho LC, Ayres VJ, Seleti RO, Ferreira CA et al (2017) Silicone-induced granuloma of breast implant capsule (SIGBIC): similarities and differences with anaplastic large cell lymphoma (ALCL) and their differential diagnosis. Breast Cancer 9:133–140
Headon H, Kasem A, Mokbel K (2015) Capsular contracture after breast augmentation: an update for clinical practice. Arch Plast Surg 42(5):532–543
Kappel RM, Boer LL, Dijkman H (2016) Gel bleed and rupture of silicone breast implants investigated by light-, electron microscopy and energy dispersive X-ray. Analysis of internal organs and nervous tissue. Clin Med Rev Case Rep 3:087
Cohen Tervaert JW, Kappel RM (2013) Silicone implant incompatibility syndrome (SIIS): a frequent cause of ASIA (Shoenfeld’s syndrome). Immunol Res 56(2–3):293–298
cbr.org.br [homepage on the internet]. São Paulo. Colégio Brasileiro de Radiologia. Available from: http://www.unimagem-net.com.br/cbrportal/pec_mama/aula.php. Accessed 18 June 2017
Conflict of interest
The authors state that there are no existing or potential conflicts of interest including financial, consultant, institutional and other relationships that might lead to bias or a conflict of interest. None of the data presented in this study have been published previously. The authors declare that this study has not received any funding. The authors guarantee that the article has been submitted only to Insights into Imaging.
About this article
Cite this article
de Faria Castro Fleury, E., Gianini, A.C., Ayres, V. et al. Breast magnetic resonance imaging: tips for the diagnosis of silicone-induced granuloma of a breast implant capsule (SIGBIC). Insights Imaging 8, 439–446 (2017). https://doi.org/10.1007/s13244-017-0564-3
- Breast implant
- Silicones; granuloma
- Magnetic resonance imaging
- Breast neoplasm